- Pallets pass in front of the X-ray beam on two levels
- Pallets can pass in front of the beam multiple times with each pallet quadrant being exposed equally
- Conveyance mechanism moves pallets in front of the beam at a speed of 0.5 to 4.5 m/min. Speed regulation is independent of the number of pallets loaded on this section of the conveyor
- The beam area or treatment area conveyor is 3 m long (2 such sections – top and bottom)
- Conveyor mechanism provides a gap closing conveyor sequence so that minimal gap between pallets is constant
The product is treated from 2 sides and the bottom pallet has to get the same double side treatment on the upper position, forming a 4 passes system. The depth of the irradiation is made on the long side of the pallets (1.2 m).
In order to give dose flexibility, the treatment is given by ~2kGy increments, corresponding to one full sequence of 4 passes in front of the X-ray target.
When multiple dose increments are required, i.e.,2, the corresponding container will travel 48 times (12 x 4) around the central wall in order to alternate the face which is presented to the X-ray. After every 2 passes it is changing the level from bottom to upper or contrary.
Radiation Dose Setting: Single Lot vs. Full Validation of Minimum Sterilization Dose
This TechTip will explain the differences between single lot validations and full validations of minimum sterilization dose, and outline the advantages and disadvantages of each of these approaches to dose setting.
Cumulative Radiation Dosing
Radiation dose is cumulative and the effect of absorbed dose on the destruction of micro-organisms is cumulative. Learn more about considerations for cumulative dosing.
Monitoring of Delivered Dose Using Reference Dosimetry
Monitoring of the minimum delivered dose position for healthcare products is required per industry standards (i.e. ANSI/AAMI/ISO 11137)