High-energy gamma radiation produces electron disruptions (ionization) in any material that it encounters. In living cells, these disruptions result in damage to the DNA and other cellular structures. These photon-induced changes at the molecular level may cause the death of the organism or render the organism incapable of reproduction. This effect is useful in killing bacteria, insects, or other living contaminants which may exist in, or on, a product.
When ionizing radiation passes through a material, its energy is deposited into the material. The energy deposited into material can be done so in order to accumulate a cumulative amount of energy, or dose. Radiation dose is cumulative and the effect of absorbed dose on the destruction of micro-organisms is cumulative. Therefore, product can be irradiated several times in order to accumulate a specific dose.
Products that are capable of supporting microbial growth may be able to support cumulative dosing. An assessment of the impact on the process interrupt between cumulative doses for these types of products should be performed by the product manufacturer.
Generally, unless the product can support growth based on some nutritive quality or component, process interruption should not be problematic. During radiation processing, when the delivered dose falls short of the products specified dose range the product can be augmented, or “topped off”, to achieve the requested dose range as radiation dose is cumulative.
Without specific knowledge of the product, in general, products that do not have high moisture content are not nutritive for bacteria and therefore their dose can be augmented by the cumulative dosing. Again, an assessment for these types of products should be performed by the product manufacturer.
AAMI TIR 29:2012. Guide for process control in radiation sterilization of medical devices
(Sections 220.127.116.11.4 and 18.104.22.168.5)
AAMI 11137-1:2006, Appendix A10.10
Fairand, Barry P., “Radiation Sterilization for Health Care Products, X-Ray, Gamma, and Electron Beam”, p.103. (2002).
Overview Of Sterilization Technology Comparison
Learn more about our global offering, which includes electron beam, gamma, X-ray, and ethylene oxide technologies
Transfer of Sterilization Dose, Verification Dose or Maximum Acceptable Dose Between Radiation Sources
This document answers many of the FAQs regarding radiation source transfers and addresses possible transfer types and areas for consideration.
Monitoring of Delivered Dose Using Reference Dosimetry
Monitoring of the minimum delivered dose position for healthcare products is required per industry standards (i.e. ANSI/AAMI/ISO 11137)